Fully Auto Microbial ID/AST System
DL-96A is fully automated microbial ID/AST system intergrated with incubator and dispensing inside.
Feature & Principle
-Fully automatically operation.
-Detection and analysis of microbial infection.
-Capacity: 60 IST cards.
-Identify around 500 kinds of microorganisms, and cover 200 kinds of antibiotics with latest CLSI regulation.
-Divide the antibiotics into 4 groups by expert analysis system, providing True MIC and SIR result.
-Fully automated closed system (sample insertion, incubation, reading of results, interpretation of results);
-Simultaneous identification of a microorganism and determination of its sensitivity to antibiotics in one panel;
-The identification of microorganisms is carried out by two methods: colorimetry and turbidimity;
-Registration of the growth of microorganisms is carried out by the device according to two parameters: Cell replication score and assessment of the metabolic activity of the cell.
-The determination of sensitivity to antibiotics should is carried out by the degree of turbidity and by changing the redox potential.
-Determination of sensitivity to antibiotics by serial microdilution method to detect the minimum inhibitory concentration of antibiotic: Definition of “delayed resistance” with prediction of the development of stability, and the results of the identification of microorganisms could be nearly 5 hours, antibiotic Sensitivity Results could be nearly 8 hours.
-Sound and visual notification of the presence of a positive sample with an indication on the display of the location of the positive sample.
-For number of identified microorganisms, there are not less than 140 Gram-positive aerobic and facultative anaerobic bacteria, and not less than 160 Gram-negative aerobic and facultative anaerobic bacteria;
-The sensitivity of microorganisms is determined by at least 25 antibacterial drugs;
-Special tests inluding extended Spectrum Beta-Lactamase (ESBL) and methicillin-resistant staphylococci (MRS).
-Saving the results of the analysis in the memory of the device for not less than 30 days.
-A nephelometer for measuring the turbidity of suspensions of microorganisms in the range of at least 0- 6.0 according to the McFarland standard with a calibration kit: at least 1 set.
Publicity materials download link：